Please wait for the process to complete.
Checking for non-preferred file/folder path names (may take a long time depending on the number of files/folders) ...
This resource contains some files/folders that have non-preferred characters in their name. Show non-conforming files/folders.
||This resource does not have an owner who is an active HydroShare user. Contact CUAHSI (email@example.com) for information on this resource.|
|Storage:||The size of this resource is 494.6 KB|
|Created:||Apr 01, 2018 at 6:29 p.m.|
|Last updated:|| Apr 09, 2018 at 6:16 p.m.
|Citation:||See how to cite this resource|
|+1 Votes:||Be the first one to this.|
|Comments:||No comments (yet)|
Hydrologic research is a very demanding application of fiber-optic distributed temperature sensing (DTS) in terms of precision, accuracy and calibration. The physics behind the most frequently used DTS instruments are considered as they apply to four calibration methods for single-ended DTS installations. The new methods presented are more accurate than the instrument-calibrated data, achieving accuracies on the order of tenths of a degree root mean square error (RMSE) and mean bias. Effects of localized non-uniformities that violate the assumptions of single-ended calibration data are explored and quantified. Experimental design considerations such as selection of integration times or selection of the length of the reference sections are discussed, and the impacts of these considerations on calibrated temperatures are explored in two case studies.
Raw project data is available by contacting firstname.lastname@example.org
How to Cite
This resource is shared under the Creative Commons Attribution CC BY.http://creativecommons.org/licenses/by/4.0/
There are currently no comments