Identifying and Correcting Step Losses in Single-Ended Fiber-Optic Distributed Temperature Sensing Data
|Authors:||Hausner, M. B.|
|Resource type:||Composite Resource|
|Created:||Mar 31, 2018 at 8:36 p.m.|
|Last updated:||Apr 09, 2018 at 8:34 p.m. by CTEMPs OSU-UNR|
Fiber-optic distributed temperature sensing (DTS) makes it possible to observe temperatures on spatial scales as fine as centimeters and at frequencies up to 1 Hz. Over the past decade, fiber-optic DTS instruments have increasingly been employed to monitor environmental temperatures, from oceans to atmospheric monitoring. Because of the nature of environmental deployments, optical fibers deployed for research purposes often encounter step losses in the Raman spectra signal. Whether these phenomena occur due to cable damage or impingements, sharp bends in the deployed cable, or connections and splices, the step losses are usually not adequately addressed by the calibration routines provided by instrument manufacturers and can be overlooked in postprocessing calibration routines as well. Here we provide a method to identify and correct for the effects of step losses in raw Raman spectra data. The utility of the correction is demonstrated with case studies, including synthetic and laboratory data sets.
Raw project data is available by contacting firstname.lastname@example.org
DTS,fiber optic,distributed temperature sensing,CTEMPs
How to cite
This resource is shared under the Creative Commons Attribution CC BY.http://creativecommons.org/licenses/by/4.0/
|Hausner, M. B.|
Select content in the file browser to see metadata specific to that content. Metadata will only display here when the the content is selected above. Content specific metadata does not display on the Discover page.
Please wait for the process to complete.