Please wait for the process to complete.
Checking for non-preferred file/folder path names (may take a long time depending on the number of files/folders) ...
This resource contains some files/folders that have non-preferred characters in their name. Show non-conforming files/folders.
||This resource does not have an owner who is an active HydroShare user. Contact CUAHSI (email@example.com) for information on this resource.|
|Storage:||The size of this resource is 298.7 KB|
|Created:||Apr 01, 2018 at 6:18 p.m.|
|Last updated:|| Apr 09, 2018 at 6:18 p.m.
|Citation:||See how to cite this resource|
|+1 Votes:||Be the first one to this.|
|Comments:||No comments (yet)|
Over the past five years, Distributed Temperature Sensing (DTS) along fiber optic cables using Raman backscattering has become an important tool in the environmental sciences. Many environmental applications of DTS demand very accurate temperature measurements, with typical RMSE < 0.1 K. The aim of this paper is to describe and clarify the advantages and disadvantages of double-ended calibration to achieve such accuracy under field conditions. By measuring backscatter from both ends of the fiber optic cable, one can redress the effects of differential attenuation, as caused by bends, splices, and connectors. The methodological principles behind the double-ended calibration are presented, together with a set of practical considerations for field deployment. The results from a field experiment are presented, which show that with double-ended calibration good accuracies can be attained in the field.
Raw project data is available by contacting firstname.lastname@example.org
How to Cite
This resource is shared under the Creative Commons Attribution CC BY.http://creativecommons.org/licenses/by/4.0/