Please wait for the process to complete.
Checking for non-preferred file/folder path names (may take a long time depending on the number of files/folders) ...
This resource contains some files/folders that have non-preferred characters in their name. Show non-conforming files/folders.
||This resource does not have an owner who is an active HydroShare user. Contact CUAHSI (firstname.lastname@example.org) for information on this resource.|
|Storage:||The size of this resource is 838.2 KB|
|Created:||Apr 01, 2018 at 6:12 p.m.|
|Last updated:|| Apr 09, 2018 at 6:24 p.m.
|Citation:||See how to cite this resource|
|+1 Votes:||Be the first one to this.|
|Comments:||No comments (yet)|
Characterizing both spatial and temporal soil moisture (θ) dynamics at site scales is difficult with existing technologies. To address this shortcoming, we developed a distributed soil moisture sensing system that employs a distributed temperature sensing system to monitor thermal response at 2 m intervals along the length of a buried cable which is subjected to heat pulses. The cable temperature response to heating, which is strongly dependent on soil moisture, was empirically related to colocated, dielectric-based θ measurements at three locations. Spatially distributed, and temporally continuous estimates of θ were obtained in dry conditions (θ ≤ 0.31) using this technology (root mean square error [RMSE] = 0.016), but insensitivity of the instrument response curve adversely affected accuracy under wet conditions (RMSE = 0.050).
Raw project data is available by contacting email@example.com
How to Cite
This resource is shared under the Creative Commons Attribution CC BY.http://creativecommons.org/licenses/by/4.0/
There are currently no comments